All rights reserved. Printed in the Netherlands. All product names are trademarks of their respective companies. Fluke // ScopeMeter. Users Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 . 2 Fluke Corporation ScopeMeter C/B Series, Series and the NEW. Path: /fluke/B/. This is the These manuals are available for download and free of charge. Fluke B/B-C/B-C ScopeMeter Service Manual.
|Published (Last):||19 February 2018|
|PDF File Size:||4.52 Mb|
|ePub File Size:||2.80 Mb|
|Price:||Free* [*Free Regsitration Required]|
Fluke scopemeter family datasheet – series ii 10 pages.
Perform an auto set. Hold down to mmanual the trigger delay. The actual reading with its A relative measurement displays the present bargraph is displayed beneath these readings.
For additional optional accessories, see the ScopeMeter Accessories msnual. Chapters Table Of Contents For other countries, a line plug adapter complying with the applicable National Requirements must be used. Chapter 1 Using The Scope About this Chapter This chapter provides a step-by-step introduction to the scope functions of the test tool.
Contents continued Triggering on Video Signals You can display a fixed reference waveform with the – the last used reference waveform actual waveform for comparison. Note that the Freezing the Screen Peak-Peak scopemmeter for input B appears next to the input A You can freeze the screen all readings and waveforms at frequency reading at the top of the screen. All previous recordings are stored in memory. Display the key labels. See Figure measurement result relative to a defined reference value.
Tips Chapter 7 Tips About this Chapter Using the Standard Accessories This chapter gives you information and tips on how you The following illustrations show the use of the standard can make the best use of the test tool.
FLUKE B USER MANUAL Pdf Download.
To activate wcopemeter ranging, do the following during any Meter measurement: Fluke t and t electrical tester datasheet 2 pages. Now, you will see a screen like in Figure 15 Figure ZOOM see the complete waveform and then return to the zoomed part.
Triggering on Waveforms Using Trigger Delay or Pre-trigger Using Trigger Delay or Pre-trigger You can begin to display the waveform some time before or after the trigger point has been detected. Apply a video signal to the red input A. In this example it is assumed that the Condition. Triggering on Missing Pulses Chapter 3 Using The Recorder Functions About this Chapter This chapter provides a step-by-step introduction to the recorder functions of the test tool.
This function optimizes the position, range, time base, and triggering and assures a stable display of virtually any waveform.
Both, the reference screen and the Display the key labels. If the signal changes, the setup is automatically adjusted to maintain the best display result.
Triggering on Waveforms Triggering scopeemeter Video Signals Triggering on Video Signals To trigger on a video signal, first select the standard of the video signal you are going to measure: Capturing Intermittents Automatically From Scope mode, open the When you use the test tool in triggered fulke, A marker is To measure rise time, do the following: Enter text from picture: Chooseaccept the new Display the key labels.
Observe also that the bargraph is displayed. Tips Using the Tilt Stand Using the Tilt Stand Resetting the Test Tool The test tool is equipped with a tilt stand, allowing viewing If you want to reset the test tool to the manaul settings, do from an angle while placed on a table. To set the trigger delay, do the following: For example, 1992b following: Read First Safety Information: Set the vertical amplitude scale to linear or logarithmic.
Start on trigger to start recording; Specified non-tolerance including approvalUL including approval Safety To replay continuously, do the following: The Spectrum function shows the spectral content of the input A or input B waveform.
It performs an FFT to transform the amplitude waveform from the time domain into the frequency domain.
Fluke 192B User Manual
The response is quicker because the test tool is be changed as follows. The display also shows the actual waveform. Page 71 Triggering on Waveforms Triggering on Pulses Finding Missing Pulses Select the positive pulse icon to trigger on the gap between the The scopmeeter example covers finding missing pulses in a train positive pulses, then jump to of positive pulses.
With the next procedure you can choose how auto set behaves when you press the auto set key. The set includes the following items not available separately: For complete service, not use abrasives, solvents, or alcohol. See next This chapter provides an introduction to the trigger section. RS, optically isolated with probe check.